These support guides will help you handle and care for your OPUS 2 Audio Processor.
When the indicator light on the control unit continuously blinks red, it’s time to replace the batteries.
Pull the battery pack cover down and remove it from the battery pack frame.
Use the magnetized underside of the D Coil to remove the used batteries from the battery pack frame. Place the coil over each battery one at a time to remove each from the frame.
Insert the new zinc-air batteries with positive (+) poles facing out. Before you insert the new battery set, make sure the battery contacts are clean and dry. Avoid touching the battery contacts inside the battery pack frame. Finally, remove the foil stickers covering the batteries. Always replace all of the batteries at once rather than mixing used and new batteries.
To turn OPUS 2 on, slide the battery pack cover onto the frame and close the battery pack lock. Don’t use excessive force.
Use only type 675 batteries to power your OPUS 2. We recommend using zinc-air batteries that are designed specifically for cochlear implant systems. When you use OPUS 2 with the WaterWear accessory, however, you must substitute zinc-air batteries with another type of battery.
| Maximum |
| Requires |
| Compatible |
|XS Battery Pack|
|675 Zinc-Air Batteries||60 hours|
|Standard Battery Pack|
|Zink-Air Batteries||90 hours|
|Silver-Oxide Batteries||10 hours|
|Alkaline Batteries||8 hours|
|Rechargable Batteries||16 hours|
Refer to your user manual for more information.